ICE FOAM is a non-toxic product that does not cause health damage. However, if swallowed, drinking plenty of water is recommended. In case of abnormal symptoms, seek medical attention.
ICE FOAM packaging should not be opened as it protects the foam.
It is necessary to keep at least 72 hours freezing.
The ICE FOAM in large quantity should be discarded in “industrial landfill”. Please contact our quality department at (11) 4341-8600.
The Thermal Blanket is a thermal insulator used at room temperature and has the benefit of avoiding initial negative temperature peaks and helps to maintain it for longer time.
The Thermal Blanket should be stored at room temperature. ICE FOAM can also be stored at room temperature, requiring freezing for 72 hours before use.
In order that FIT BOX obtains desired and effective thermal maintenance, it is necessary to use ice and thermal insulation.
SUPER COLD contains an additive that changes the melting point of ice to below 0 ° C and is suitable for transporting frozen products.
Tech Gel does not cause health damage. However, if swallowed, drinking plenty of water is recommended. In case of abnormal symptoms, seek medical attention.
The Tech Gel bottle should not be opened. If it is opened, the gel contained inside it will be expelled.
It is necessary to leave at least 72 hours freezing.
The Tech Gel in large quantity should be discarded in “industrial landfill”. Polar Técnica offers to dispose of the product to its customers.
ENVIRONMENTS: It is recommended the annual qualification in the summer and winter seasons.
EQUIPMENT: It is recommended that the qualification be carried out annually. However, if the equipment undergoes any repair or critical alteration, it must be requalified.
All devices are calibrated (RBC traceable) and the software is validated according to ANVISA Computerized System Validation Guide or FDA 21 CFR Part 11.
The objective is to verify with temperature records the homogeneity and the thermal stability of the environment and, based on the records, identify the most critical points, the hottest and the coldest points, for the decision making with regard to the points to be monitored in the routine.
Installation Qualification (IQ): Step in which it is verified that the equipment conforms to the manufacturer’s manual and is correctly installed in relation to the electrical network. In case of environment, check the same data of the equipment that operates in the place for the maintenance of the temperature.
Operation Qualification (OQ): Step where the correct operation of the equipment is verified, the set-point and the on / off function to verify its reaction when these commands are executed.
Performance Qualification (PQ): Stage where the equipment or environment undergoes a study of at least 24 hours of thermal mapping, being 24 hours without load and 24 hours with load, when possible. For the warehouse it is recommended a minimum of 72 hours of thermal mapping.
At the end of the thermal mapping study, the door opening and power outage tests were performed to test the performance of the equipment under these conditions and verify its stability with temperature variation.